|MODEL||Plum A+ Triple|
8" H x 8" W x 6" D, excluding pole clamp
9.5 lbs. with battery
High-impact plastic casing
120V, 50-60 Hz, 35VA
F1, F2, 250V, 0.5A fuses
Hospital Grade AC Cord 10 ft long
One sealed lead acid, rechargeable 6V battery, internal
With a new fully charged battery, the infuser shall operate for a minimum of 6 hours at 125 ml/hr or less, or deliver 500 ml at 126 ml/hr or greater on one line
The battery charges whenever the pump is connected to AC power. The recharge time is approximately 6 hours with the device operating at 125 ml/hr on one line
Delivery Rate Range:
Lines A & B:
0.1-99.9 ml/hr in 0.1 ml increments 100-999 ml/hr in 1 ml increments
Concurrent Delivery: 0.5 ml/hr minimum for each line
Plum Set: 500 ml/hr cumulative (A+B) maximum
KVO: 1.0 ml/hr or the last primary delivery rate, whichever is less
0.1-99.9 ml in 0.1 ml increments 100-9999 ml in 1 ml increments
Turn to Run
VTBI Complete or Dose End
Alarm Pressure: 1-15 psig (default 6 psig)
Speak with a KWIPPED representative at (800) 273-8404 (9-5pm ET), email [email protected] or chat with us below:
KWIPPED offers unparalleled efficiency to source equipment rentals and leases from a global network of equipment suppliers.
Tell us where to send your quotes.
Eventually, we'll need a few more details to facilitate your rental/finance, but we know your time is valuable.
Do you want to submit the request now, or do you have some time to enter company details?
Upon letting us know the information below, you will have full access to KWIPPED's platform.